ASP-150. Multipurpose Spectrometer
|Spectral range for choosing the registration domain*, nm||190-1100||190-1100||190-1100||190-1100||190-1100||190-800||190-450||190-300|
|Registered spectrum region width*(1), nm||950||640||480||315||145||90||40||25|
|Spectral resolution*, nm||0.6||0.42||0.3||0.2||0.09||0.06||0.025||0.017|
|Input slit||10 µm|
|Focal length||150 mm|
|Focal length of the camera objective||150 mm|
|Input type(2)||free-space and FC fiber socket (fiber patch-cord on request)|
|Dimensions (LxWxH)||242х115х76 mm (ASP-150T); 205х115х76 mm (ASP-150C)|
|CCD array specifications(3)|
|Model||Toshiba 1205DG||Toshiba 1304DG||Hamamatsu
|Number of pixels||2048||3648||1024|
|Pixel width||14 µm||8 µm||25 µm|
|Pixel height||200 µm||200 µm||500 µm|
|Minimum exposure time||4.2 ms||7.3 ms||2.1 ms|
|Maximum exposure time||4 s||5 s||5 s|
|Meansquare noise of reading, counts of ADC||5.4||3.5||16(4.4)(5)|
|Dynamic range||1000:1||1000:1||1000:1 (4000:1)(5)|
|(1) - the necessary width and center of the registered range should be specified when placing an order. The ASP-150T has an operational range adjustment function;
(2) - each spectrometer is equipped with a fiber socket and allows measurement of either free-space or fiber signals without any realignment;
(3) - any array can be combined with any grating type;
(4) - sensitivity given at the wavelength of 550 nm;
(5) - the Hamamatsu arrays provide for sensitivity control via special driving signal, that can set either low (values in brackets) or high sensitivity mode;
(6) - the property of the CCD to prevent the charge flow from the neighboring overexposed pixels;
* - values given for the Toshiba 1304DG CCD array.
The ASP-150C spectrometer is an ideal choice for a wide variety of different optical applications with its flexible tuning and high resolution. There is always a place for the ASP-150C on the optical table due to its small size, functionality and friendly interface.
The ASP-150T model is able to carry out measurements in a wider spectral range without any losses in resolution, owing to a special tunable mechanism.
All models have a special fiber input with a spectral slit, the size of which is conditioned with the detector. It allows measurement of either free-space or fiber signals without any realignment. However, it is not recommended to use fiber to measure spectrum of a femtosecond laser due to signal modulation and subsequent spectrum distortion.
Dear colleagues, Avesta will be upgrading this corporate web-site starting 04 Nov 2017. Please note that you might experience some turbulence while browsing the web-site in the process. We hope to finish the upgrade as soon as possible to introduce new looks, updated products and more information.
Avesta is participating in the “LASER World of PHOTONICS 2017” (Munich, Germany, 26-29 June 2017) trade show. Feel free to drop by our booth B2.151 to see the TeMa Yb laser system along with the AA-20 autocorrelator and GECON supercontinuum generator. The trade show participation is supported by the Russian Export Center and organized by Expocenter and Russian Laser Association.